|Home||» Products||» Hospital Nebulizer Machine||» Hospital Nebulizer Machine|
General Safety Precautions:
Provide close supervision when this Hospital Nebulizer Machine is used by,on,or near children or in valids. Inspect the device and parts before using the them each time,and check that there are no problems. In particular,be sure to check the following:
a) Make sure that the baffle,air filter,and other parts are attached correctly. Also check that the baffle,nozzle or air tube are not damaged.
b) Make sure that the air filter is clean.If the air filter has changed colour,or has been- That the nozzle is not blocked. used on average for more than 60 days, replace it with a new one. So that the compressor operates normally.
c) Do not tilt the Hospital Nebulizer Machine so that it is at an angle of greater than 45Â°, or shake it.
d) When using this device,there will be some noise and vibration caused by the pump in.
e) while in use. the compressor.There will also be some noise caused by the emission of compressed.
f) Do not use or store the device while the air tube is bent.air from the Hospital Nebulizer Machine. This is normal and does not indicate a malfunction,or Approved for human use only.reduce the working life of the device.
g) Use only original parts and accessories. Use of unsupported parts may damage Operate the device only as intended. Do not use the device for any other purpose.
h) Do not use the device at temperatures greater than+40Â°C.
i) Do not add more than 7 ml of medication to the medication cup.Dispose of the device and any used accessories or optional parts according to.
j) Do not drop the compressor or Hospital Nebulizer Machine,or subject them to strong shocks.applicable local regulations.Unlawful disposal may cause environmental pollution.
k) Do not distort the baffle,or jab the nozzle of the medication cup with a pin etc.
l) Do not insert fingers or objects inside the compressor.
m) Do not block the air filter cover.
n) Do not disassemble or attempt to repair the compressor or power cord.
0) To avoid injury to the nose mucosa,do not squeeze the nosepiece into the back of the nose.
p) Do not block the slit between the cap and the inhalation air inlet.
q) Do not use a microwave oven or hair dryer to dry the nebulizing parts.
r) Changes or modifications not approved by will void the user warranty.